The Freedom™ CPAP Battery gives you the freedom to power your PAP device when and where you need it - no matter where your world takes you!
Travel: The perfect solution for remaining compliant with your PAP therapy when enjoying your favorite outdoor activities, taking adventures on the road, or riding the high seas. Meets FAA requirements for lithium-ion batteries taken in carry-on baggage during air travel.
Power Outages: Never worry about sleepless nights without your PAP therapy during power outages. Simply plug the Freedom™ CPAP Battery into the wall outlet and connect your PAP device to the battery. When AC power is available it will bypass the battery cells to power your device but during an outage the battery will recognize the loss of AC power and automatically begin powering your PAP device.
Other Devices: As an added bonus the USB port and/or included DC input cord let you use the Freedom™ CPAP Battery to charge other small electronic devices such as cell phones and tablets.
PAP DEVICE COMPATIBILITY
ResMed S8, Fisher & Paykel ICON, Transcend, 3B Medical, and Apex Medical PAP series devices via the included DC input cord and 150W Sine Wave Power Inverter.
Respironics System One, Respironics M Series, and Devilbiss IntelliPAP series devices via the included connector tips and DC power pigtail cord.
Note: Some older obsolete PAP devices and targeted ventilation PAP devices may not be compatible or may require more power than the Freedom™ CPAP Battery can provide.
ESTIMATED RUN TIMES:
ResMed S8, Fisher & Paykel ICON, 3B Medical, and Apex Medical PAP devices: 1 night (without the use of heated humidification and heated tubing) before needing to be charged.
Transcend devices: 1-2 nights (without the use of heated humidification and heated tubing) before needing to be charged.
Respironics System One, Respironics M Series and Devilbiss IntelliPAP series devices: 1-3 nights (without the use of heated humidification and heated tubing) before needing to be charged.
Run times are estimates and depend upon several factors including PAP device type, pressure setting, and humidifier usage. All run times are based on CPAP devices without the use of heated humidification or heated tubing on a pressure setting of 10 cm/H2O.
The use of heated humidification and heated tubing will reduce run times by two-thirds in most cases and in some cases could prevent the battery from being able to power the PAP device. For optimal CPAP Battery performance you should plan to use your machine without heated humidification or heated tubing.
BiPAP and AutoPAP machines will have lower run times as the pressure setting is not constant and changes throughout the night requiring more power to operate the machine.
FREEDOM™ BATTERY SPECIFICATIONS
- Cell Type: Industrial grade rechargeable lithium-ion
- Capacity: 97.68 Wh (per battery)
- Output Voltage Range: 9V-12.6V DC
- USB Output Voltage: 5V
- Output Current: 8A maximum
- Power Gauge: Digital display charge level indicator
- Charge Time: 4 hours (per battery)
- Size: 7.5 x 4.8 x 0.8 (L x W x H) (per battery)
- Weight: 1.7 lbs (per battery)
- Casing: Aluminum
- Warranty: 1 year
150W SINE WAVE POWER INVERTER SPECIFICATIONS
- Input: 12V DC
- Output: 110-120V AC
- Output Rated Power: 150W (200W maximum)
- USB Output: 5V DC, 1A/2.1A
- Dimensions: 3.5' x 2.25' x 1' (L x W x H) | Cord Length: 31.5'
- Weight: 6.2 oz
- Warranty: 1 year